You can download this for your platform and open the application and go to tools serial monitor. that will listen to the configured serial port. (You do not need an arduino for it to work.)
From their website:
Q: How do I use Hyperterminal to check communications?
A: Hyper Terminal is a Terminal program that is bundled with Windows.
To set up Hyper Terminal for use with an Ohaus balance, perform the following steps. The following assumptions are made for the purpose of explanation.
1. The computer uses Com1
2. The Balance Baud Rate is set to 2400
3. The Balance Data Bits are set to 7
4. The Balance Stop Bits are set to 2
5. The Balance Parity Bit is set to None
6. Handshaking in the balance is turned off
Open the program and enter a name for the connection. You will probably want to save the setup for future use.
In the next screen click on the twirl down for the "Connect using" box and select the COM port your computer uses.
Click OK and the COMx properties box will be displayed. (Which COM port appears will depend on the setting in the previous step.)
Set the parameters to match the parameters in the balance. See the instruction manual for the balance. The usual defaults are 2400 baud, 7 data bits, no parity, 2 stop bits and no handshake. When finished press apply then press OK.
The terminal screen will be displayed.
Select CALL from the menu bar and select Disconnect. The "Connected" at the bottom left of the screen will change to "Disconnected".
Select FILE from the menu bar and select Properties.
Select the Settings tab.
Check Windows Keys and Del.
At the bottom of the box click on ASCII Setup.
Check the first two boxes and put 11 in the next 2.
Click Ok twice.
From the Call menu select Call.
"Call" will be displayed at the lower left of the terminal screen.
You are now ready to communicate with the balance. The best way to test the connection is to press print on the balance. Other commands are listed in the instruction manual for the balance.
And some additional info from that site:
How can I get my Defender 5000 indicator (T51P or T51XW) to print without inserting an extra line feed? I am using a label printer with preprinted labels and every other label is wasted.
A: You can do this by going into the menu under Print1 or Print2 and setting the format to Single.
Q: Ok now I am printing a single label but there is too much information. I am still using a T51P.
A: Again go into the Print1 or Print 2 menu. You can turn off all the heading and GLP information you need too. You can also turn off the Gross, Net and Tare printing and just leave the result. You also have the option to print the result as a number only by setting it to NUM. This will eliminate the unit of measure.
Q: I am using a 7000 series Defender indicator (T17P) and I can not clear the accumulator. I have followed the instructions in the manual but it does not clear.
A: Press and hold the "info" button until the display shows "StatS". Then press the "CLR" button to clear the accumulator.
taken from: http://us.ohaus.com/us/en/home/support/faq.aspx#S16